ITEM Condition and TESTS PERFORMED. The item is in used condition unless a different condition is noted in the ITEM DESCRIPTION.
Please see ITEM DESCRIPTION for tests performed. Items shown in the pictures and listed in the description are included.
IMPORTANT INFORMATION - PLEASE READ. We are not familiar with every item we sell. Some items may have missing parts or accessories and we do not realize they are missing. Do NOT assume parts or accessories are included.
If they are not in the photos or in the description, then they are not included. Please review the description and photos carefully.2396 The degasser powers on. The pumps runs, after about one minute the pump turns off and the yellow light in the upper right corner goes out. We did not test with any liquids or samples.
We have not tested further. The serial number on the back does not match the serial number on the front. Item is bubble wrapped and secured in box with packing materials. We do our best to describe the items we sell. We will work with you to resolve the issue as we want to have satisfied customers.The item "Agilent HP 1100 Series G1322A Degasser, Tested, Working" is in sale since Tuesday, October 25, 2016. This item is in the category "Business & Industrial\Healthcare, Lab & Life Science\Lab Equipment\Analytical Instruments".
The seller is "labinstruments80" and is located in Granite Quarry, North Carolina. This item can be shipped to United States, to Canada, to United Kingdom, DK, RO, SK, BG, CZ, FI, HU, LV, LT, MT, EE, to Australia, GR, PT, CY, SI, to Japan, to China, SE, KR, ID, to Taiwan, to Belgium, to France, to Hong Kong, to Ireland, to Netherlands, PL, to Spain, to Italy, to Germany, to Austria, IL, to New Zealand, SG, to Switzerland, NO, SA, UA, AE, QA, KW, BH, HR, MY, CO, PA, JM, BB, BD, BM, BN, BO, EC, EG, GF, GG, GI, GP, IS, JE, JO, KH, KY, LI, LK, LU, MC, MO, MQ, MV, NI, OM, PE, PK, PY, RE.