This item is for a brand new portable stainless steel pressure steam sterilizer is one of best in the same grade sterilizer. It uses saturated steam to rapidly and efficiently sterilize articles, It can be used in hospitals, public health centers, health stations, clinics in factories and mines, scientific research institutes, etc, to sterilize medical equipment surgical dressings, glass utensils, solutions, substrata, etc.It can also be used as cooker in plasteau areas and be used to make high quality drinking water in industrial and mining enterprises. Used to sterilize medical equipment, surgical dressings, glass utensils.
Quickly rises temperature and low power consumption. Working pressure:0.14-0.16Mpa.Max working pressure:0.165Mpa. Voltage:AC110V for united states/canada/Mexico. AC220V for Europe or other country. Power of the Electro-Thermal Tube:2kw. Sources of ignition: Electric heat. Instruction for Timing Adjustment Model. Ensure proper operation of the sterilizer, it must be placed on horizontal surface.
Open the device cover, take out the sterilized indoor tube (heating electric heating tube), and pour 3.5 liters of water into the electric heating tube. After each time use, you must replenish the right amount of water in time. In case of burning the electric heating tube, due to lack of water. Item are stacked in the disinfection chamber in turn, there should be a gap between the stacking bag and the package to facilitate steam penetration, in order to improve the disinfection effect.
The safety valvesteam outlet of the steam release valve should not be blocked. Finally, close the lid and tighten the screws diagonally. Verify that the power supply specifications for the equipment.Then plugging in the power supply. Plug in, adjust the timer that you needed, (clockwise to increase, maximum timing is 60 minutes), then adjust the temperature as well.
(maximum temperature is 126 °C). After setting, turn on the power switch.
At this time, the heating indicator lights up on the panel , the device is working. During process, when the temperature reaches the set temperature, it automatically enters the heat preservation and enter next stage to the sterilization. When it is done, it will sound beep. You can unplug the device.
In 2 to 3 minutes later, the exhaust valve is vented and the pressure gauge shows zero, you can open device finally. Customer service is from Monday to Friday, all the message will be replied in 12 hours.
A: Please log on your account. Q: How do I return an item? Please contact our customer service before you return an item. The item "18/24L Stainless Steel Electric Autoclave Sterilizer Dental Medical Equipment" is in sale since Monday, March 18, 2019. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Medical & Lab Equipment, Devices\Autoclaves & Sterilizers".
The seller is "litattoo8" and is located in BEIJING. This item can be shipped worldwide.